Upper Central Incisors and upper Canines
10 mm / 12 mm / 14 mm
ZrO2 – zirconium oxide – zirconia – (Y-TZP)
The diameter of the implant is 4.1 mm in the apical area and 4.8 mm in the most coronal threaded part. The transgingival part is 6 mm wide.
MAXIMUM TORQUE = 35N
This is a ONE-PIECE implant (monoblock). The space between the occlusal part of the abutment and the antagonist should be around 1.5 mm. The abutment is designed to avoid grinding once the implant is inserted into the socket. It has an abutment ready to restore. The gum heals around the uncovered abutment during the 3-4 months after surgery.